- Stock: In Stock
- Brand: Teledyne Isco
- Product Code: 52244
- Condition: Pre-Owned
Here is a Teledyne Isco CombiFlash NextGen 300+ Chromatography System
This Teledyne Isco CombiFlash NextGen 300+ Chromatography System is in good working condition.
Please see exact photos and video below
The testing we performed:
The Teledyne Isco CombiFlash NextGen 300 Chromatography System passed all tests and was found to be fully functional:
Tested Power and Initialization: PASSED
Tested Touch Screen Calibration: PASSED
Tested Arm Movement: PASSED
Tested RFID Column Detection: PASSED
Tested Waste and Source Level Sensing: PASSED
Tested Manual Control Function: PASSED
Tested Prime Function: PASSED
Tested Gradient: PASSED
Tested UV Detection of Sodium Iodide: PASSED
The system consists of:
Teledyne Isco CombiFlash NextGen 300+ Chromatography System
1 x Collection Rack of your choice (let us know so we can match it to the collection tubes you are using)
Intake hoses with caps and solvent filters (Qty.:4)
Set of waste hoses
We carry on hand a variety of racks and other accessories; pleas inquire with us for availability.
Configuration and Setup:
Flow Rate Range: 1–300 mL/min
Maximum Pressure: up to 20 bar (300 psi); user defined pressure limits
UV Detection: Variable UV 200–400 nm
Number of Solvents: binary gradient, (selecting two solvents from four inlets and also allows for the addition of a third solvent modifier.)
Gradient Formation: low pressure
Programmable Gradients: linear, step, isocratic
Gradient Accuracy: ±2% of full scale
Dimensions (H x W x D): 26 x 14.1 x 17 in (66 x 36 x 43 cm)
Weight: 61 lb (27.7 kg)
Power Requirements: 100-240VAC, 50/60Hz, 2A
The video below gives an overview of the Teledyne Isco CombiFlash NextGen 300+ Chromatography System, and also demonstrates its functionality.
Datasheet: Teledyne Isco CombiFlash NextGen 300+ Chromatography System
Brochure: Teledyne Isco CombiFlash NextGen 300+ Chromatography System
Manual: Teledyne Isco CombiFlash NextGen 300+ Chromatography System
This will ship via freight. Please contact us with your zip code to obtain shipping quote.
*SM* Small Crate
Please do not hesitate to contact us if you have any questions.
All equipment warranty request must have a support case opened within the 60-day warranty period.
To open a support case, please send an email to [email protected] or call (508)974-4896.
Please provide your name, contact information, quotation/invoice number and a brief description of the issue that you are experiencing.
Phone and/or Email support will be provided prior to scheduling a return.
Please see the link below for our detailed warranty statement.
Customers must retain all original shipping materials during the warranty period.
Every instrument has unique packing and shipping needs. Few instruments can be safely shipped via Pacel, and most require the safety of a well-built crate.
Here we are showing you our standard packing process
1. All items are different, but they all follow the same principle when packaging.
2. Start with a well-constructed crate
A well-constructed crate would consist of 1/2” to 5/8” Plywood with framing. We generally purchase our crates from a third party who specializes in custom crates.
3. You always want the Instrument to not be directly on the wood deck of the crate. This is done by placing it on foam, inside of the crate. We use 2” 1.7 lb. density foam which works great. It provides cushioning and enough firmness to support the Instrument.
4. The next step is to make sure there is nothing on the deck of the Instrument and the Head/ Arm is secured. In some cases there will be multiple arms to secure but for this example there is only one. The reason for this is so there is no free movement during transit which could cause serious damage to the Instrument.
5. The next step is to make sure your Instrument is covered in either a pallet bag or shrink wrap so no dust or foreign object will get into the crate.
6. The next step is to secure the instrument inside the crate by bracing. We use a method of 2x4’s which have 1” foam shrink wrapped to them so no piece of wood touches the actual instrument.
7. Final step is to photograph everything inside the crate and then seal it with screws.