- Stock: In Stock
- Brand: Agilent
- Product Code: 50408
- Condition: Pre-Owned
Here is an Agilent Bravo Automated Liquid Handler with Multi TEC Control and Thermo Cube
System pre-setup for NGS
- This system is equipped with a 96 multichannel head that allows for a liquid transfer up to 250ul
- This system is able to perform plate to plate transfers
- This system is able to perform reservoir to plate transfers
- The included gripper allows the labware to be moved around the deck
- This is a disposable tip system to prevent cross-contamination
- This system will be correctly packed for safe shipping either domestically and internationally
- The magnetic separation position accommodates 96 well PCR plates
- The Thermo Cube chilled position allows for chilling down to 1 degree C with distilled water
- The included component combination was decided based on Agilent G5573AA NGS Option A site prep checklist
- The PlateLoc Sealer and/or On Deck Incubator options can be included for an additional charge
This system is a great choice for nucleic acid preparation and enrichment prior to sequencing.
This system has the hardware needed to run NGS methods for Agilent SureSelect, Agilent HaloPlex, Illumina TruSeq, NuGEN Ovation, KAPA HTP, PacBio SMRTbell, and NEBNext
The system is designed to ensure consistency in sequencing processes. This standardization can be particularly valuable for users who may not have in-depth technical knowledge, as it minimizes the risk of errors and ensures reliable results.
This system is an all-in-one solution for NGS, incorporating various tools and functionalities needed for the entire sequencing workflow. This can be advantageous for users who prefer a packaged solution over piecing together individual components.
This Agilent Bravo Automated Liquid Handler with Multi TEC Control and Thermo Cube is in good working condition
Please see the exact video and photos below.
Testing we performed:
This Agilent Bravo Automated Liquid Handler 16050-102 passed all tests and was found to be fully functional:
Power on: PASSED
Liquid transfer using 96-head: PASSED
Plate transfer using Bravo Gripper Arm: PASSED
Heater 1 and Heater 2: PASSED
Emergency Stop button: PASSED
Light Curtain: PASSED
Thermo Chube Chiller: PASSED
Inheco MTC: PASSED
System consists of:
Agilent Bravo Liquid Handler
96LT 96-well disposable-tip pipette head
Bravo Gripper upgrade
Deck platepad (standard Bravo, Qty.: 4)
Alignment station platepad (passive, 3-springed rollers)
Heating platepad (Qty.: 2)
Inheco MTC Controller
Teleshake Module platepad
Thermo Scientific Shakemodule
Thermal Station (cooling pad)
Thermo Cube Solid State Cooling System
Magnetic Bead Accessory
Robot Disable Hub G5266A
Bravo front Light Curtain w/ side splash shield
Emergency Stop Button (Qty. 2)
Computer running VWorks software v14.2
Communication and power cables
Configuration and Setup:
Robot Arm(s): 96LT head, Bravo Gripper upgrade
Channels: 96 (max 250 per channel)
Usable Tips: 10, 30, 70, 200, 250
Microplates: 96- or 384-well formats
Application: liquid handling, and sample preparation
Customizable deck postions (Qty. 9)
The video below gives an overview of the Agilent Bravo Automated Liquid Handler with Multi TEC Control and Thermo Cube, and demonstrates its functionality.
Brochure: Agilent Bravo Liquid Handler
Manual: Agilent Bravo Liquid Handler
Manual: Multi TEC Control
Manual: Thermo Cube
This will ship via freight. Please contact us with your zip code to obtain shipping quote.
*LA* Large Crate
Please do not hesitate to contact us if you have any questions.
All equipment warranty request must have a support case opened within the 60-day warranty period.
To open a support case, please send an email to [email protected] or call (508)974-4896.
Please provide your name, contact information, quotation/invoice number and a brief description of the issue that you are experiencing.
Phone and/or Email support will be provided prior to scheduling a return.
Please see the link below for our detailed warranty statement.
Customers must retain all original shipping materials during the warranty period.