- Stock: In Stock
- Brand: Peak
- Product Code: 31335
Here is a Peak NM20ZA LC-MS Nitrogen Generator
Please make sure it`s output is sufficient for your instrument. Please check with us if unsure.
This Peak NM20ZA Nitrogen Generator is in good working order.
Please see the exact video and photos below.
Testing we performed:
The Peak NM20ZA Nitrogen Generator passed all tests and was found to be fully functional:
Tested System Initialization: PASSED
Tested Compressor Check: PASSED
Tested Curtain Nitrogen % to 97%: PASSED
Tested Curtain Maximum Flow: PASSED
Tested Source Maximum Flow: PASSED
Tested Exhaust Maximum Flow: PASSED
System consists of:
Peak NM20ZA Nitrogen Generator
Standard power cord.
Configuration and Setup:
Maximum Relative Humidity: 70%
Maximum Operating Ambient Temperature: 25C (77F)
Power: 230V AC +/-10% (50/60 Hz) 8.4 Amps
Curtain Maximum Flow: 12 l/min @ 5.52 bar (0.42 CFM @ 80 PSI)
Net Weight: 145 kg (319 lb)
Minimum operating ambient temperature 5C (41F)
Maximum Altitude 2000 Meters
Source Maximum Flow 26 l/min @ 7.6 bar (0.92 CFM @ 110 PSI)
Exhaust Maximum Flow 8.0 l/min @ 4.2 bar (0.29 CFM @ 60 PSI)
Particles < 0.01 um
Outlets 1/4" BSP 3
Pressure Gauges 3
Drains Ports 1/4" BSP 2
Suspended Liquids NONE
Compressor fuse on Control PCB 8.0 amps
PLC / fan fuse on Control PCB 2.0 amps
Electrical connection Single Phase Power Cord
Noise Level 57 dBA @ 1m
The video gives an overview of the Peak NM20ZA Nitrogen Generator, and also performs functionality.
This will ship via freight. Please contact us with your zip code to obtain shipping quote.
*ME* Medium Crate
Please do not hesitate to contact us if you have any questions.
Here we are showing you our standard packing process
1. All items are different, but they all follow the same principle when packaging.
2. Start with a well-constructed crate
A well-constructed crate would consist of 1/2” to 5/8” Plywood with framing. We generally purchase our crates from a third party who specializes in custom crates.
3. You always want the Instrument to not be directly on the wood deck of the crate. This is done by placing it on foam, inside of the crate. We use 2” 1.7 lb. density foam which works great. It provides cushioning and enough firmness to support the Instrument.
4. The next step is to make sure there is nothing on the deck of the Instrument and the Head/ Arm is secured. In some cases there will be multiple arms to secure but for this example there is only one. The reason for this is so there is no free movement during transit which could cause serious damage to the Instrument.
5. The next step is to make sure your Instrument is covered in either a pallet bag or shrink wrap so no dust or foreign object will get into the crate.
6. The next step is to secure the instrument inside the crate by bracing. We use a method of 2x4’s which have 1” foam shrink wrapped to them so no piece of wood touches the actual instrument.
7. Final step is to photograph everything inside the crate and then seal it with screws.
All equipment warranty request must have a support case opened within the 60-day Domestic USA and 30-day International warranty period.
To open a support case, please send an email to [email protected] or call (508)974-4896.
Please provide your name, contact information, quotation/invoice number and a brief description of the issue that you are experiencing.
Phone and/or Email support will be provided prior to scheduling a return.
Please see the link below for our detailed warranty statement.
Customers must retain all original shipping materials during the warranty period.