- Stock: In Stock
- Brand: Olympus
- Product Code: 25558
Here is an Olympus IX81 Fluorescence Motorized Microscope IX81F-3 Pred IX83
Setup for Brightfield with Fluorescence studies.
This microscope is optimized for observation of biological studies.
The LEP motorized stage has a travel range of 120 x 100mm. The custom stage insert will accommodate slides, petri dish, or wellplates.
The total magnification range of this microscope is 40-600x
This Olympus IX81 Fluorescence Motorized Microscope is in good working order with all aspects tested to be working fully.
All optics of this microscope have been cleaned professionally under a class 10,000 clean room hood.
Please see the exact video and photos below.
Testing we performed:
The Olympus IX81 Fluorescence Motorized Microscope passed all tests and was found to be fully functional:
Tested halogen lamp and source fully: PASSED
Tested fluorescence and source fully: PASSED
Tested transmission halogen light path: PASSED
Tested reflective fluorescence light path: PASSED
Tested objectives for clarity: PASSED
Tested port-splitter to split between camera and eyepieces: PASSED
Tested mechanical condenser: PASSED
Tested Z-axis: PASSED
Tested motorized fluorescence turret: PASSED
Tested motorized objective turret virtually and with controller: PASSED
Tested connection from camera to laptop: PASSED
Tested control of motorized stage virtually and with joystick: PASSED
Tested virtual control of the microscope: PASSED
Tested live feed for clarity: PASSED
Tested all filter slides: PASSED
Tested all motorized components including Objective Turret, Z-Axis, XY stage, Brightfield Illumination intensity, Fluorescence Turret, Port switching: PASSED
During testing this Olympus IX81 Fluorescence Motorized Microscope was found to have no issues.
A clear image was produced through the eyepieces as well as through the camera port.
All of the mechanical and Motorized components moved freely and the test sample was able to be focused without excessive effort.
System consists of:
Olympus IX81 Inverted Microscope
IX2-UCB Power/Control Box w/ remote controller
U-BI90 Binocular Head
WHN10X/22 and WHN10X-H/22 Eyepieces
Prisms / Condensers:
IX-LWCD Mechanical Condenser NA0.55
Brightfield Light Path:
IX2-ILL100 Illumination Tower
Fluorescence Light Path:
IX2-RFACA Motorized 6-position Turret
Aqua (Blue) - EX 436/20, DM 455, EM 480/30 (#F0777)
FITC (Green) - EX 475/42, DM 505, EM 460/50 (#F0793)
CY3 (Red) - EX 535/50, DM 565, EM 620/60 (#F0493)
12V 100W Halogen Lamp w/ IX2-UCB power supply
CRI (Prior Lumen200) 200W Metal Halide Lamp w/ light guide and collimating adapter
LEP 99S906 Motorized XY Stage
LEP MAC5000 Control box
LEP XY Joystick
Motorized 6-position Objective Turret
4x/0.13 UPlanFl (#O2552)
10x/0.30 U PlanFL (#O2496)
40x/0.65 Ach Ph2 (#O1248)
Photo / Port Split:
Left 100% or Head 100%
AmScope MU1403 14.0MP CMOS USB3.0 Digital Camera
Computer and Software:
Dell Latitude i5 running AmScope and Olympus BSW software and MicroManager software
Objective Turret, Z-Axis, XY stage, Brightfield Illumination intensity, Fluorescence Turret, Port switching
Communication and power cables.
Configuration and Setup:
Illumination: Brightfield w/ Fluorescence
Objectives: 4x/0.13 UPlanFl, 10x/0.30 U PlanFL, 40x/0.65 Ach Ph2
Light Sources: 12V 100W Halogen Lamp, CRI (Prior Lumen200) 200W Metal Halide Lamp
Stage: Motorized XY
This Olympus IX81 Fluorescence Motorized Microscope is currently set up for Brightfield with Fluorescence studies.
The AmScope CMOS digital colored camera is capable of taking detailed digital images.
This Olympus IX81 Fluorescence Motorized Microscope system includes a Dell Latitude i5 to control the AmScope software, and Olympus BSW software and MicroManager software. .
The video starts with a front view of the Olympus IX81 Fluorescence Motorized Microscope.
After that we perform functionality of all components of the Olympus IX81 Fluorescence Motorized Microscope system.
The video ends with another wide front view of the Olympus IX81 Fluorescence Motorized Microscope.
This will ship via freight. Please contact us with your zip code to obtain shipping quote.
Please do not hesitate to contact us if you have any questions.
Here we are showing you our standard packing process
1. All items are different, but they all follow the same principle when packaging.
2. Start with a well-constructed crate
A well-constructed crate would consist of 1/2” to 5/8” Plywood with framing. We generally purchase our crates from a third party who specializes in custom crates.
3. You always want the Instrument to not be directly on the wood deck of the crate. This is done by placing it on foam, inside of the crate. We use 2” 1.7 lb. density foam which works great. It provides cushioning and enough firmness to support the Instrument.
4. The next step is to make sure there is nothing on the deck of the Instrument and the Head/ Arm is secured. In some cases there will be multiple arms to secure but for this example there is only one. The reason for this is so there is no free movement during transit which could cause serious damage to the Instrument.
5. The next step is to make sure your Instrument is covered in either a pallet bag or shrink wrap so no dust or foreign object will get into the crate.
6. The next step is to secure the instrument inside the crate by bracing. We use a method of 2x4’s which have 1” foam shrink wrapped to them so no piece of wood touches the actual instrument.
7. Final step is to photograph everything inside the crate and then seal it with screws.
All equipment warranty request must have a support case opened within the 60-day Domestic USA and 30-day International warranty period.
To open a support case, please send an email to [email protected] or call (508)974-4896.
Please provide your name, contact information, quotation/invoice number and a brief description of the issue that you are experiencing.
Phone and/or Email support will be provided prior to scheduling a return.
Please see the link below for our detailed warranty statement.
Customers must retain all original shipping materials during the warranty period.