- Stock: In Stock
- Brand: Nikon
- Product Code: 18153
Here is a Nikon TE2000-E PFS Perfect Focus Inverted Fluorescence Motorized Phase Contrast DIC Microscope Pred Ti/TI2
Setup for Brighfield with Fluorescence, DIC and Phase Contrast studies.
This microscope is optimized for microplate or standard slide sample observation that requires enhanced contrasting options, as well as observation stability in multiple magnification ranges.
The addition of the PFS (Perfect Focus System) allow for near effortless swapping between samples of different focal ranges
Ti or TI2 series
This Nikon TE2000-E PFS Inverted Fluorescence Motorized Phase Contrast DIC Microscope is in good working order with all aspects of this unit tested and working fully.
Microscope optics have been cleaned professionally by our technician under a class 10,000 clean room hood.
Please see the exact video and photos below.
Testing we performed:
The Nikon TE2000-E PFS Inverted Fluorescence Motorized Phase Contrast DIC Microscope passed all tests and was found to be fully functional:
Tested halogen lamp and source fully: PASSED
Tested metal halide lamp and light path: PASSED
Tested transmission halogen light path: PASSED
Tested reflective fluorescence light path: PASSED
Tested objectives for clarity: PASSED
Tested port-splitter to split between camera and eyepieces: PASSED
Tested mechanical condenser: PASSED
Tested motorized Z-axis: PASSED
Tested motorized fluorescence turret: PASSED
Tested motorized objective turret: PASSED
Tested connection from camera to laptop: PASSED
Tested motorized stage with joystick: PASSED
Tested live feed for clarity: PASSED
Tested all filter slides: PASSED
Tested connection and functionality for control of the microscope through Nikon TE software and MicroManager: PASSED
Tested all motorized parts including XY Stage, Z-axis, autofocus, objective turret, filter cube turret, halogen light intensity: PASSED
During testing and cleaning this Nikon TE2000-E PFS Inverted Fluorescence Motorized Phase Contrast DIC Microscope was found to have no issues.
A clear image was produced through the eyepieces as well as through the camera port.
All of the mechanical components moved freely and the test sample was able to be focused without excessive effort.
System consists of:
Nikon TE2000-E Inverted Microscope
T-HUBC HUB Controller
T-RCP Remote Control Pad
T-EFN Focus Knob
T-BSUK 70 Stage Riser
4-position Lens Turret - Closed (C), Open (O), Bertrand (B) and 2.5x Magnification (M)
(2) CFI 10x/22 Eyepieces
LWD 0.52 Mechanical Condenser
N2 Insert (#DIC-0242)
Ph2 Prism (#PH-0119)
Fluorescence Light Path:
Fluorescence Filter Cubes:
T-FLC-E 6-position Motorized Turret
DAPI (Blue): EX 377/50, DM 409, EM 447/60 (#F00214)
B-2E/C (FITC/Green): EX 480/30, DM 505, DM 535/40 (#F00198)
TRITC HYQ (Red): EX 545/30, DM 570, EM 620/60 (#F00310)
LHS-H100P-1 12V 100W Halogen Lamp w/ TE2-PS100W power supply
EXFO X-Cite eXacte XCT10A 200W Metal Halide Light Source w/ C-HGFIB Culminating Adapter and light guide (New in Box, with new lamp)
LEP Motorized Stage Model 99S908-BC
LEP MAC5000 Stage Controller
LEP Joystick Model MAC2 XY
T-ND6-E 6-position Motorized Turret
4x/0.2 Plan Apo (#O2054)
10x/0.30 Plan Fluor DIC L (#O2055)
20x/0.75 Plan Apo DIC N2 (#O0199)
40x/0.65 LWD Ph2 DL (#O1545)
T-PFS Perfect Focus System
T-PFC Perfect Focus Controller
Analyzer / Polarizer:
T-A2 Analyzer (#DIC-0189)
DIC N2 Insert (#DIC-0242)
Photo / Port Splits:
100% Left Port
100% Front Port
100% Bottom Port
50/50 Eye/Right Port
AmScope MU1003 10MP Digital Camera
Dell Latitude i5 running AmScope software, Nikon TE software and MicroManager software
XY Stage, Z-axis, autofocus, objective turret, filter cube turret, halogen light intensity
Communication and power cables.
Configuration and Setup:
Illumination Type: Brighfield w/ Fluorescence, Phase Contrast, DIC
Objectives: 4x/0.2 Plan Apo, 10x/0.30 Plan Fluor DIC L, 20x/0.75 Plan Apo DIC N2, 40x/0.65 LWD Ph2 DL
Light Source: 12V 100W Halogen, EXFO X-Cite 200W Metal Halide
Stage: Motorized XY
This Nikon TE2000-E PFS Inverted Fluorescence Motorized Phase Contrast DIC Microscope is currently set up for Brighfield with Fluorescence, DIC and Phase Contrast studies.
The MU1003 10MP AmScope CMOS digital color camera is capable of taking detailed digital images.
The Nikon TE2000-E PFS Inverted Fluorescence Motorized Phase Contrast DIC Microscope system includes a Dell Latitude i5 laptop to control the AmScope software, Nikon TE software and MicroManager software.
The video gives a walkaround of the Nikon TE2000-E PFS Inverted Fluorescence Motorized Phase Contrast DIC Microscope, showing overall condition and including serial tags.
The video shows functionality of the Nikon TE2000-E PFS Inverted Fluorescence Motorized Phase Contrast DIC Microscope.
This will ship via freight. Please contact us with your zip code to obtain shipping quote.
Please do not hesitate to contact us if you have any questions.
Here we are showing you our standard packing process
1. All items are different, but they all follow the same principle when packaging.
2. Start with a well-constructed crate
A well-constructed crate would consist of 1/2” to 5/8” Plywood with framing. We generally purchase our crates from a third party who specializes in custom crates.
3. You always want the Instrument to not be directly on the wood deck of the crate. This is done by placing it on foam, inside of the crate. We use 2” 1.7 lb. density foam which works great. It provides cushioning and enough firmness to support the Instrument.
4. The next step is to make sure there is nothing on the deck of the Instrument and the Head/ Arm is secured. In some cases there will be multiple arms to secure but for this example there is only one. The reason for this is so there is no free movement during transit which could cause serious damage to the Instrument.
5. The next step is to make sure your Instrument is covered in either a pallet bag or shrink wrap so no dust or foreign object will get into the crate.
6. The next step is to secure the instrument inside the crate by bracing. We use a method of 2x4’s which have 1” foam shrink wrapped to them so no piece of wood touches the actual instrument.
7. Final step is to photograph everything inside the crate and then seal it with screws.
All equipment warranty request must have a support case opened within the 60-day Domestic USA and 30-day International warranty period.
To open a support case, please send an email to firstname.lastname@example.org or call (508)974-4896.
Please provide your name, contact information, quotation/invoice number and a brief description of the issue that you are experiencing.
Phone and/or Email support will be provided prior to scheduling a return.
Please see the link below for our detailed warranty statement.
Customers must retain all original shipping materials during the warranty period.