- Stock: In Stock
- Brand: Biotek Bio-Tek
- Product Code: 31479
Here is a BioTek ELx405 Select Microplate Washer ELX405UVS w/ Valve Module Pred 405 TS/LS
Current model is 405 TS/LS
This BioTek ELx405 Select Microplate Washer is in good working order.
Please see the exact video and photos below.
Testing we performed:
The BioTek ELx405 Select Microplate Washer passed all tests and found to be fully functional:
Tested initial power boot system: PASSED
Tested control panel buttons and display: PASSED
Tested method creation: PASSED
Tested prime and wash cycles: PASSED
Tested dispense and aspiration: PASSED
Tested plate movement: PASSED
Tested soak and rinse cycles: PASSED
Tested autoclean with ultra sonic cleaner function: PASSED
System consists of:
BioTek ELx405 Select Microplate Washer ELX405UVS
Dual Action Manifold (for 96 & 384 Well plates)
Vacuum Pump (Pump provided may be different than pump shown)
Power cords, and hoses
Configuration and Setup:
Usable Labware: 96 & 384 well Microplates
Manifold Channels: 96 for aspiration and 96 for dispense
Number of Dispensing Liquids: 4
Dispense Volume Range: 50 to 3000 uL/well, in 1-9 uL increments
Options: Dual Manifold, Valve Module, Built-in Ultra Sonic Cleaner
"U" - 96-/384-well microplate washing: 96-tube (8 x 12) manifold + Dual-Action
"V" - automatic 4 buffer switching (Valve Module can be added upon request)
"S" - built-in ultrasonic cleaner for manifold self-maintenance
The video gives an overview of the BioTek ELx405 Well Microplate Washer, showing overall condition, and then performs functionality.
25" x 25" x 25" @ 45#, 21" x 17" x 15" @ 30#
Please do not hesitate to contact us if you have any questions.
Here we are showing you our standard packing process
1. All items are different, but they all follow the same principle when packaging.
2. Start with a well-constructed crate
A well-constructed crate would consist of 1/2” to 5/8” Plywood with framing. We generally purchase our crates from a third party who specializes in custom crates.
3. You always want the Instrument to not be directly on the wood deck of the crate. This is done by placing it on foam, inside of the crate. We use 2” 1.7 lb. density foam which works great. It provides cushioning and enough firmness to support the Instrument.
4. The next step is to make sure there is nothing on the deck of the Instrument and the Head/ Arm is secured. In some cases there will be multiple arms to secure but for this example there is only one. The reason for this is so there is no free movement during transit which could cause serious damage to the Instrument.
5. The next step is to make sure your Instrument is covered in either a pallet bag or shrink wrap so no dust or foreign object will get into the crate.
6. The next step is to secure the instrument inside the crate by bracing. We use a method of 2x4’s which have 1” foam shrink wrapped to them so no piece of wood touches the actual instrument.
7. Final step is to photograph everything inside the crate and then seal it with screws.
All equipment warranty request must have a support case opened within the 60-day Domestic USA and 30-day International warranty period.
To open a support case, please send an email to [email protected] or call (508)974-4896.
Please provide your name, contact information, quotation/invoice number and a brief description of the issue that you are experiencing.
Phone and/or Email support will be provided prior to scheduling a return.
Please see the link below for our detailed warranty statement.
Customers must retain all original shipping materials during the warranty period.