- Stock: In Stock
- Brand: Beckman Coulter
- Product Code: 30518
Here is a Beckman Biomek 4000 Liquid Handler A99749
Last manufacturer PM 3/12/21
- This liquid handler is capable of serial dilutions, reservoir to plate, and plate to plate transfers
- The liquid transfer can be 1 or 8 disposable tip at a tip depending on needed transfer
- The maximum volume that can be transferred at one time is 200ul
- This liquid handler is intended to be used with standard and deepwell microplates
- There are 9 open positions for microplates or tipboxes on the deck
- Multiple tools can be used in a single method for more complex liquid transfers
New costs over $60K
This Beckman Biomek 4000 Liquid Handler is in good working order.
There are minor cracks to the plastic enclosure that do not impede functionality of the liquid handler or the enclosure.
Please see the exact video and photos below.
Testing we performed:
The Beckman Biomek 4000 Liquid Handler passed all tests and was found to be fully functional:
Tested initialization: PASSED
Tested AccuFrame teaching tool: PASSED
Tested pipette tool loading: PASSED
Tested aspiration: PASSED
Tested dispense: PASSED
Tested gripper: PASSED
System consists of:
Beckman Coulter Biomek 4000 Liquid Handler
AccuFrame teaching position
Desktop PC running Biomek software ver4.1 (w/ keyboard, monitor, mouse)
Communication and power cables
Configuration and Setup:
Robotic Arm(s): interchangeable tool arm w/ (4) pipette heads
Channels: 1 or 8
Volume: 10 to 200ul
Tips Type: disposable
Application: liquid handling and sample preparation
The video starts with an overview of the Beckman Biomek 4000 Liquid Handler.
After that we observe functionality of the Beckman Biomek 4000 Liquid Handler system.
This will ship via freight. Please contact us with your zip code to obtain shipping quote.
*LO* Long Crate
OB Packed Weight - 500lb
Please do not hesitate to contact us if you have any questions.
Here we are showing you our standard packing process
1. All items are different, but they all follow the same principle when packaging.
2. Start with a well-constructed crate
A well-constructed crate would consist of 1/2” to 5/8” Plywood with framing. We generally purchase our crates from a third party who specializes in custom crates.
3. You always want the Instrument to not be directly on the wood deck of the crate. This is done by placing it on foam, inside of the crate. We use 2” 1.7 lb. density foam which works great. It provides cushioning and enough firmness to support the Instrument.
4. The next step is to make sure there is nothing on the deck of the Instrument and the Head/ Arm is secured. In some cases there will be multiple arms to secure but for this example there is only one. The reason for this is so there is no free movement during transit which could cause serious damage to the Instrument.
5. The next step is to make sure your Instrument is covered in either a pallet bag or shrink wrap so no dust or foreign object will get into the crate.
6. The next step is to secure the instrument inside the crate by bracing. We use a method of 2x4’s which have 1” foam shrink wrapped to them so no piece of wood touches the actual instrument.
7. Final step is to photograph everything inside the crate and then seal it with screws.
All equipment warranty request must have a support case opened within the 60-day Domestic USA and 30-day International warranty period.
To open a support case, please send an email to [email protected] or call (508)974-4896.
Please provide your name, contact information, quotation/invoice number and a brief description of the issue that you are experiencing.
Phone and/or Email support will be provided prior to scheduling a return.
Please see the link below for our detailed warranty statement.
Customers must retain all original shipping materials during the warranty period.