- Stock: In Stock
- Brand: Agilent
- Product Code: 30720
Here is an Agilent SureScan G4900DA G2600D Microarray Scanner
New costs about $160K
This Agilent SureScan G4900DA G2600D Microarray Scanner is in good working order.
Please see the exact video and photos below.
Testing we performed:
The Agilent SureScan G4900DA G2600D Microarray Scanner passed all tests and found to be fully functional:
Tested main power: PASSED
Tested communication with software: PASSED
Completed Agilent Scanner Self Test: PASSED
Scan of Test Slide: PASSED
Performed scan of test slide and found to be operating fully. Recommend PM to ensure best performance.
System consists of:
Agilent SureScan Microarray Scanner G2600D
Computer running Scan Control software ver9.1
Communication and power cables.
Configuration and Setup:
Dynamic Range: >10^4 (16-bit data format), >10^5 (20-bit data format), 10^6 (with XDR scanning)
Resolution: 2, 3, 5 and 10 microns
Autoloader: 24-slide cassette allows for hands-off operation
Laser Information: Green solid-state laser, 532nm; Red solid-state laser, 640nm
Scan Window Maximum: 71 x 21.6mm
The video gives an overview of the Agilent SureScan G4900DA G2600D Microarray Scanner and demonstrates communication with included Scan Control software.
This will ship via freight. Please contact us with your zip code to obtain shipping quote.
*SM* Small Crate
Please do not hesitate to contact us if you have any questions.
Here we are showing you our standard packing process
1. All items are different, but they all follow the same principle when packaging.
2. Start with a well-constructed crate
A well-constructed crate would consist of 1/2” to 5/8” Plywood with framing. We generally purchase our crates from a third party who specializes in custom crates.
3. You always want the Instrument to not be directly on the wood deck of the crate. This is done by placing it on foam, inside of the crate. We use 2” 1.7 lb. density foam which works great. It provides cushioning and enough firmness to support the Instrument.
4. The next step is to make sure there is nothing on the deck of the Instrument and the Head/ Arm is secured. In some cases there will be multiple arms to secure but for this example there is only one. The reason for this is so there is no free movement during transit which could cause serious damage to the Instrument.
5. The next step is to make sure your Instrument is covered in either a pallet bag or shrink wrap so no dust or foreign object will get into the crate.
6. The next step is to secure the instrument inside the crate by bracing. We use a method of 2x4’s which have 1” foam shrink wrapped to them so no piece of wood touches the actual instrument.
7. Final step is to photograph everything inside the crate and then seal it with screws.
All equipment warranty request must have a support case opened within the 60-day Domestic USA and 30-day International warranty period.
To open a support case, please send an email to [email protected] or call (508)974-4896.
Please provide your name, contact information, quotation/invoice number and a brief description of the issue that you are experiencing.
Phone and/or Email support will be provided prior to scheduling a return.
Please see the link below for our detailed warranty statement.
Customers must retain all original shipping materials during the warranty period.