- Stock: SOLD
- Brand: Advanced Instruments
- Product Code: 18624
Here is an Advanced Instruments 3250 Osmometer
New costs about $18K
Comes with a new kit of Osmolality Linearity Set and New Clinitrol Reference Solution
This Advanced Instruments 3250 Osmometer is in good working order.
Please see the exact video and photos below.
Testing we performed:
The Advanced Instruments 3250 Osmometer passed all tests and found to be fully functional:
Tested power switch and keypad function: PASSED
Tested and passed initial diagnostics: PASSED
Tested and passed head up/down test: PASSED
Tested and passed stir/freeze test: PASSED
Tested and passed probe bin test: PASSED
Tested and passed barcode test: PASSED
Tested and passed display/printer test: PASSED
Tested keypad buttons: PASSED
System consists of:
Advanced Instruments 3250 Single Sample Osmometer
New Clinitrol 290 Reference Solution 10x5 mL
New Osmolality Linearity Set 100-2000 mOsm/kg, 10x5 mL
New Heat Transfer Fluid 1x150 mL
Probe Alignment tool
Configuration and Setup:
Osmolality Reading: low range 0-2000mOsm; high range 1500-4000mOsm
Printer built into unit
Sample Capacity: 1 sample
Sample Size: 0.20-0.25mL
Test Time: low range 2 minutes, high range 3 minutes
RS-232 Serial Port
Optional Barcode Scanner (NOT INCLUDED)
Dimensions (WxDxH): 13 x 18 x 16in
Operating Temperature: 18 to 35 C
Chamber Cooling: 1 to -8C
The video starts in front of the Advanced Instruments 3250 Osmometer and shows that it is on and ready to function. A sample is inserted and a run is started.
The video then rotates left than right to the rear and shows all warranty stickers and tags of the Advanced Instruments 3250 Osmometer.
The video then shows the consumables that will be included with it and the interior of the Advanced Instruments 3250 Osmometer.
The video then waits for the run to end with a favorable result. The video pans out and ends.
Please do not hesitate to contact us if you have any questions.
Here we are showing you our standard packing process
1. All items are different, but they all follow the same principle when packaging.
2. Start with a well-constructed crate
A well-constructed crate would consist of 1/2” to 5/8” Plywood with framing. We generally purchase our crates from a third party who specializes in custom crates.
3. You always want the Instrument to not be directly on the wood deck of the crate. This is done by placing it on foam, inside of the crate. We use 2” 1.7 lb. density foam which works great. It provides cushioning and enough firmness to support the Instrument.
4. The next step is to make sure there is nothing on the deck of the Instrument and the Head/ Arm is secured. In some cases there will be multiple arms to secure but for this example there is only one. The reason for this is so there is no free movement during transit which could cause serious damage to the Instrument.
5. The next step is to make sure your Instrument is covered in either a pallet bag or shrink wrap so no dust or foreign object will get into the crate.
6. The next step is to secure the instrument inside the crate by bracing. We use a method of 2x4’s which have 1” foam shrink wrapped to them so no piece of wood touches the actual instrument.
7. Final step is to photograph everything inside the crate and then seal it with screws.
All equipment warranty request must have a support case opened within the 60-day Domestic USA and 30-day International warranty period.
To open a support case, please send an email to firstname.lastname@example.org or call (508)974-4896.
Please provide your name, contact information, quotation/invoice number and a brief description of the issue that you are experiencing.
Phone and/or Email support will be provided prior to scheduling a return.
Please see the link below for our detailed warranty statement.
Customers must retain all original shipping materials during the warranty period.